BIOTEK SYNERGY HT MANUAL PDF
The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Genre:||Health and Food|
|Published (Last):||18 December 2005|
|PDF File Size:||8.45 Mb|
|ePub File Size:||19.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
Microplates should be perfectly clean and free of dust or bottom scratches.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Page 67 Tip Prime before dispensing. Decontaminate The Fluid Lines Put the tubes back in.
Cleaning The Priming Plate As shown below, both injectors are positioned directly above the bottom fluorescence optical probe. Contact Information outside the U. A monochromator is used for absorbance measurements. Electromagnetic Interference And Susceptibility Page 6 Run the Absorbance Plate Test The installed syringes should resemble the following: Required Materials BioTek offers a liquid test kit PN containing the microplate and solutions used manyal this procedure.
Warranty And Product Registration Test Plate Certificates Biotke too much pressure can push the lens out of its holder; reinstallation must be performed by BioTek service personnel.
Computer Control The Synergy HT is completely computer-controlled; it is not equipped with a keypad or its own user interface. Place the tip priming trough in bioyek left-rear pocket of the carrier. Alternate Test Solutions Locate and grasp the handles. Page 42 Chapter 2: Chapters Table Of Contents Run the Absorbance Plate Test Refer to the software documentation for further instructions.
Saturation Errors Instrument Qualification Chapter 4 Instrument Qualification This chapter contains procedures for qualifying the initial and ongoing performance of the Synergy HT and the external dispense module if used. Stream water from a faucet through the pipe to be sure it is clean.
If the water does not stream out, try soaking the heads in hot soapy water and then reinserting the stylus. Defining the Procedure within a Gen5 Protocol Figure Remove both of them and set them aside. Remove the Snyergy Carrier Shippin To print the manual completely, please, download it.
BioTek Synergy HT Manuals
If sunergy suspect that problems occurred during shipment, if you received the reader back If you have not already done so: Absorbance Plate Test PN to confirm the mechanical alignment; optical density accuracy, linearity, and repeatability; and wavelength accuracy of the Synergy HT.
Disconnect the heater and thermistor wires. If prompted to select a reader, select the Synergy HT and click Select a plate type. Gently lift the left side of the incubator housing and carefully slide it out.
Do not block any air vents. Cleaning The Optical Probes cover. Test Solutions Chapter 4: See the photos on the next page. To reinstall a filter wheel: By pressing ‘print’ button you will print only current page.
Clean both sides of the filter with lens paper. Unpack and Inspect the Dispense Modul Page This error indicates that no autocal data for any of the three read locations bottom probe, top probe, absorbance.
Gently pull the optical probe up manuak out of its socket to expose it for cleaning. Insert the accessories box alongside the dispense module box.
Table of contents Table Of Contents Recommendations For Optimum Performance Appendix C Error Codes This section lists and describes the possible error codes that may appear on the controlling PC.
Absorbance Reader Noise Errors a16 Page – Alternate Procedure for Models with Inje This error indicates that during a self-test, the lamp did not indicate that it was off. Grasp both sides of the shroud, slide it toward you, and pull it straight off the instrument. Connect the Host Computer Page 48 –